The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Service Guide. Publication Number (order as manual set) The Agilent Technologies A is a 20 MHz synthesized function. Buy Keysight Technologies A Function Generator 20MHz GPIB, LAN, USB A. Operating manual, service manual, quick reference guide, IntuiLink.
|Published (Last):||5 July 2009|
|PDF File Size:||7.30 Mb|
|ePub File Size:||19.95 Mb|
|Price:||Free* [*Free Regsitration Required]|
This chapter describes basic signal-generation concepts and gives specific details on the internal operations of the function generator.
For example, the following command string guarantees that the first trigger is accepted and the operation is executed before the second trigger is recognized. Interface Configuration Commands including the key.
33220A Function / Arbitrary Waveform Generator, 20 MHz
The following steps show you how to configure a ms pulse waveform with a pulse width of 10 ms and edge times of 50 ns. You can issue triggers for sweeps or bursts using internal triggering, external triggering, or manual triggering.
Table of Contents Add to my manuals Add. List the names of all waveforms currently available for selection. Duty cycle represents the amount of time per cycle that the square wave is at a high level assuming that the waveform polarity is not inverted.
However, only one label for example, Freq or Period is displayed for each softkey at one time. Used in manul internal triggered burst mode only.
Agilent Technologies 33220A User Manual
If you set the carrier to a value less than the deviation frequency with FM enabledthe function generator will automatically adjust the deviation to the maximum value allowed with the present carrier frequency. Chapter 8 Specifications Specifications Page Chapter 3 Features and Functions Burst Mode Waveform Frequency The waveform frequency defines the repetition rate of the burst waveform in the triggered and external gated modes.
In order to prevent overheating, do not block the flow of air into or out of the instrument. For example, if you select a period using the PULS: MAX selects the largest dc offset for the selected function and manuak. These parameters are similar to the start frequency and stop frequency see the previous page and are included to give you added flexibility. Press the Hop Freq softkey and then set the value to Hz using the numeric keypad or the knob and cursor keys.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Page 78 Chapter 3 Features and Functions Amplitude Modulation AM Modulation Depth The modulation depth is expressed as a percentage and represents the extent of the amplitude variation. Symmetry represents the amount of time per cycle that the ramp wave is rising assuming that the waveform is not inverted.
However, you cannot overwrite any of the five built-in waveforms. Press the IP Setup softkey. Page Chapter 5 Error Messages Execution Errors Data out of range; large period limits minimum pulse width For a pulse waveform, the minimum pulse width is 20 ns for periods up to 10 seconds. Manuall can delete the waveform in volatile memory or any of the four user-defined waveforms in non-volatile memory.
Waveform Polarity In the normal mode defaultthe waveform goes positive during the first part of the cycle. The Questionable Data enable register is cleared when: Page Chapter 7 Tutorial Direct Mabual Synthesis The A represents amplitude values by 16, discrete voltage levels or bit vertical resolution.
Sending a text message to the display overrides the display state as set by the DISP command. In the normal mode defaultthe waveform goes positive during the first part of the cycle. The subnet mask indicates the portion of the host address to be used to indicate the subnet.
The remainder manuak this chapter contains examples of using the front-panel menus. The following command turns off the front-panel display. For example, the following steps show you how to convert the amplitude from Vrms to Vpp. The following steps show you how to change the amplitude to 50 mVrms. Simplified Programming Overview Chapter 4 Remote Interface Reference Simplified Programming Overview Simplified Programming Overview This section gives an overview of the basic techniques used to program the function generator over the remote interface.
Press the Stop softkey and then set the value to 5 kHz using the numeric keypad or the knob and cursor keys. The Standard Event enable register is cleared when: Chapter 5 Error Messages Execution Errors Data out of range; ramp frequency; value clipped to upper limit This generic message indicates that the waveform frequency has been limited to an upper boundary due to the selection of the ramp waveform function APPL: The function generator can produce a burst using sine, square, ramp, pulse, or arbitrary waveforms noise is allowed only in the gated burst mode and dc is not allowed.
Don’t show me this message again. All of the standard output functions except dc and noise have an associated Sync signal. For example, the following command manuwl the output frequency to 5 kHz: You mankal also control the phase offset from the front panel or over the remote interface. Chapter 3 Features and Functions Output Configuration Output Frequency As shown below, the output frequency range depends on the function currently selected. Factory Default Settings The table on the following page summarizes the factory default settings for the Agilent A.
This is a read-only register and bits are not cleared when you read the register. The GPIB address can take any value between 0 and The duty cycle deviation is expressed as a percentage of the period. You must insert a blank space to separate a parameter from a command keyword.
A Service Manual – A Waveform Generator from Agilent/Keysight
The security code is stored in non-volatile memory. A temporary message is displayed on the front panel.
Chapter 3 Features and Functions Triggering External Triggering In the external trigger mode, the function generator will accept a hardware trigger applied to the rear-panel Trig In connector.