Fluke A Multimeter DIGITAL | Business & Industrial, Test, Measurement & Inspection, Test Meters & Detectors | eBay!. The listing you’re looking for has ended. Fluke A Multimeter. Condition: For parts or not working. Ended: Jun 08, , AM PDT. Price: US $ Corporation (Fluke) has reproduced this copy of a manual which is no All shipments of JOHN FLUKE MFG. CO. . A A rechargeable Ni Cad battery pack.
|Genre:||Health and Food|
|Published (Last):||16 February 2008|
|PDF File Size:||12.31 Mb|
|ePub File Size:||16.17 Mb|
|Price:||Free* [*Free Regsitration Required]|
C22, VDC function, 2 range, 2. The oscilloscope display should be similar to waveform A1 in Figure Delivery time is estimated using our proprietary method which is based on the buyer’s proximity to the item location, the shipping service selected, the seller’s shipping history, and other factors.
Full text of “FLUKE a”
The Model 80F division ratio should be checked every month or every year depending on the accuracy requirement. When replacement is required, use only a AGX 2 amp fuse as a replacement.
Email required Address never made public. In the event the part you order has been replaced by a new or improved part, the replacement will be accompanied by an explanatory note and installation instructions, if necessary. Skip to main content. You are commenting using your Facebook account.
Note the resistance value indicated on the A display. Loosen the two screws securing the circuit board to the top of the front panel.
Q10 gate, VDC function, 2 range, 1. NOTE If the amplitude of the first ms of either oscilloscope display significantly deviates from waveform A3 or B3, U2 is the probable cause of the malfunction. Flkue the external trigger level lead to U26 pin 8. The strobe signals allow only one display LED to light at a time.
John Fluke Model A Digital Multimeter | eBay
Report item – opens in a new window or tab. Four nickel cadmium Ni-Cad batteries allow flue 8-hours of portable operation before they require recharging. This section contains an illustrated parts breakdown of the instrument.
The maximum allowable input overload condition for each function and range is given in Table The components in the A RMS Converter that perform the various functions in calculating the rms value of the input are illustrated in Figure See Figure for the oscilloscope connection points on the A boards.
Check the hoodie, which can be pulled out, in case you are doing measurements in bright light or outdoors, it will help you to see the display properly.
The Modelas shown in Figureis a clamp-on current transformer which is used to extend the current measurement capabilities of the A. The probable area in the A circuitry causing a malfunction that affects the INT signal may be indicated by the action of the signal under the two input conditions.
More chances to get what you want Feedback on our suggestions – More chances to get what you want. Remove the fuse from the mA jack press in, turn counterclockwise l A turn to release.
Table provides the gain and Buffer output information for each range. Turn R88 clockwise slowly until the A display changes by 1 or 2 digits. DlukeZevenheuvelenweg 53, Tilburg, Netherlands.
The oscilloscope should display waveform A. I used one of my own little 5V adapter. All of the digitizing process the actual counting of the oscillator pulses during the read period takes place in the custom integrated circuit U The small size and battery power add to the portability; yet the fold-away stand allows the instrument to be positioned at a convenient angle for bench top use.
Fluke 8040A Handheld “Bench” Multimeter Repair/Restore
I liked the form factor and the bench meter appeal, but still it could be handheld. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees. Remove all inputs from the A V-fl input.
The oscilloscope display shall be similar to waveform B2.