The TEG® Hemostasis Analyzer System provides a more complete .. TEG are trademarks or registered trademarks of Haemonetics Corporation in the. Consult HAEMONETICS’s entire TEG ® catalogue on MedicalExpo. Page: 1/8. Consult HAEMONETICS’s entire TEG Hemostasis Analyzer System catalogue on MedicalExpo. Page: 1/8.
|Published (Last):||14 April 2018|
|PDF File Size:||16.65 Mb|
|ePub File Size:||13.28 Mb|
|Price:||Free* [*Free Regsitration Required]|
The logon icon is then disabled in the Main toolbar. You can customize this by clicking on any of the column headings to sort on a column. On the Main Menu, click on File, then on Open. This chapter also describes the use of the case summary report. This option haenonetics controlled by the Site Administrator, and can be turned off if your site does not use these CPT codes.
Black background is useful when the captured tracing will be used on a mm slide or in an online slide presentation such as PowerPoint. It is not used to calculate the age that is manually entered there. This parameter was proposed by Raby2,3.
Splash screen It disappears in a second or two, or you can click on it or press any key to clear it. A report options wizard walks you through the needed information to generate the case summary report.
Return the lever to the load position and then press down to the eject position. Note that the identifying information in the top of the data panel is color coded to the tracing. The Main screen is displayed and contains only the records, if any, that meet your criteria.
D Use caution when performing this operation! As part of this option, you can also export the tracing coordinates to a file that can be used to plot the points in another software package.
The angle is more comprehensive than K-time, since haemonrtics are hypocoagulable haemonettics in which the final level of clot firmness does not reach an amplitude of 20 mm in which case K is undefined.
Despite the many components entering into hemostasis, two facts override all others: Local toolbar The toolbar at the top of the display panel local toolbar provides options for changing the size of tracings displayed Multi and Max as well as for viewing filtered subsets of the database Filter, Active, SiteID, and Patient and displaying the graphic of the clot Data.
They are similar in that they rely on the loss of clot strength with time after the maximum clot strength MA is reached: A unique property of this network structure is that it behaves as a rigid elastic solid, capable of resisting the deforming shear stress of flowing blood1. The exported data provides enhanced reporting and analytical capabilities when used in systems designed for such purposes. If your hard drive is C: This is the point at which most traditional coagulation assays reach their end-points.
If no dialog appears, from the File menu select either New or Open, as appropriate. The rotation movement of the pin is converted by a mechanical-electrical transducer to an electrical signal which can be monitored by a computer.
Create case screen Patient ID and patient name are required; the remaining fields are optional. If multiple analyzers are being run simultaneously, this step is done on gaemonetics channels of one analyzer before proceeding to the next. The rectangle represents the area under the curve if there had been no fibrinolysis.
TEG Hemostasis Analyzer System – HAEMONETICS – PDF Catalogs | Technical Documentation
This introduction outlines some of the various analytical techniques that can provide additional information on haemonstics blood sample. Notice that which tests appear here, as well as the order in which they are shown, is controlled by your user profile, and can be changed as described in Chapter 9. Note the text at the top of the window identifying it as a member of the user-defined library.
Click OK when you have set the options you prefer. Your user name can be entered in either upper or lower case, but the password, if used, is case sensitive.
Clusters Page 75 Case Management Chapter 5: Setting column 1 temperature 2. When a sample ends either manually or by program settingshaemoneticz is removed from the TEG screen. Barcode scanning is also available for the Operator ID field. For clinicians, operators, researchers: Notice that the button name changes to Done. See the Version 2 manual for an explanation of the files that comprise a database.
Rachwal The Toledo Hospital: To access Guide, select a tracing and then click on the Guide icon in the Main toolbar: Therefore, it appears likely that proper treatment such as platelets will normalize R as well as MA. First, however, you should check to see whether it has been excluded from the listing.
When the disposable cup pops up, lift it out of the cupwell tg dispose of it properly. This type of sensitivity does not mean this sample type is the most practical, and we will see how more practical techniques can provide similar results. Clicking the Data icon again clears the Data panel.
TEG® Thrombelastograph® Hemostasis Analyzer System – Haemonetics®
Hemostasis analyzer system Providing fast, actionable results to help you reduce risks, complications and costs. Select Operator ID and password.
The tracing border, the sample ID information, and numerical data receive a cyan blue background. Open the catalog to page 4.
TEG 5000 Thrombelastograph Hemostasis Analyzer System
This sample type was renamed in version 3 to FF—Functional Fibrinogen Level, to reflect that this sample type displays a value for functional fibrinogen level in the maximized screen. Time, rate, strength, and stability of clot indicate whether the patient has normal, hypo- or hypercoagulable hemostasis, and provide an indication of treatment necessary to normalize it. The criteria are cumulative, so that all the tabs in combination are used to filter the data.